RLC LOADS 0,3  1  2  3  5  8  10  15  20kW

VARIABLE LOAD
R  L  C
Test benches for resistive, inductive and capacitive loads
Each test bench consists of three separate load bank sections, a resistive, an inductive and a capacitive one
(RLC,) each one can be used in single phase or three phase connection, singularly or connected to other sections. Each section can dissipate its own power.
It is possible to compose mixed loads RL, RC, LC, balanced and RLC unbalanced, singlephase or threephase. Starting from
3kW a fan cooling is provided for the resistive section.
With three independent adjustment switch, one for each phase, for balanced and unbalanced steps.
MOD.x020RLC
With 7 uniform steps; 3phase (star and delta). AC monophase provides 21 regulation steps.
MOD.x02010RLC With 10 uniform steps; 3phase (star and delta). AC monophase provides 30 regulation steps.
MOD.x02015RLC With 12 loads, for 15 uniform steps; 3phase (star and delta). AC monophase provides 45 regulation steps.
MOD.x02020RLC With 20 uniform steps, (from 5% to 100%); 3phase (star & delta). AC monophase provides 60 regulation steps.
MOD.x02030RLC With 30 uniform steps, (from 3% to 100%); 3phase (star & delta). AC monophase provides 90 regulation steps.
MOD.x0201010RLC With 110 uniform steps, (from 1% to 100%); 3phase (star & delta). AC monophase provides 330 regulation steps.
With a single adjustment switch switch,
for synchronous variation of the three phases for each step.
MOD.x021RLC
With 20 steps (5%10%15%20%25%30% up to 100%) Monophase or Threephase.
MOD.x02140RLC
With 40 steps (2,5%5%7,5%10%12,5% up to 100%) Monophase or Threephase.
MOD.x02180RLC
With 80 steps (1,25%2,5%3,75%5%6,25%7,5% up to 100%) Monophase or Threephase.
On request
are available
other configurations
of adjustment steps
(load curve) or other maximum voltage and maximum power.
 Tabletop or wheeled version available.
 Optional instruments on request (Wattmeter, Voltmeter , Ammeter etc.)
ADJUSTABLE INDUCTIVE LOADS  Monophase – Threephase
With three independent adjustment switch, one for each phase, for balanced and unbalanced steps.
MOD.x020L
Each phase can be loaded with 7 independent load values
plus the zero position. In singlephase connection is possible to work with 21
different steps of regulation.
It is possible to build
balanced and unbalanced loads, singlephase or threephase Voltage: 220/380V (other voltages and powers upon request).
MOD.x02010L With 10 uniform steps; 3phase (star and delta). AC monophase provides 30 regulation steps.
MOD.x02015L With 12 loads, for 15 uniform steps; 3phase (star and delta). AC monophase provides 45 regulation steps.
MOD.x02020L With 20 uniform steps, (from 5% to 100%); 3phase (star & delta). AC monophase provides 60 regulation steps.
MOD.x02030L With 30 uniform steps, (from 3% to 100%); 3phase (star & delta). AC monophase provides 90 regulation steps.
MOD.x0201010L With 110 uniform steps, (from 1% to 100%); 3phase (star & delta). AC monophase provides 330 regulation steps.
With a single adjustment switch switch,
for synchronous variation of the three phases for each step.
MOD.x021L
Like Mod.X020L but provides 20 steps (5%10%15%20%25%30% up to 100%) Monophase or Threephase.
MOD.x02140L
Like Mod.X020L but provides 40 steps (2,5%5%7,5%10%12,5% up to 100%).
MOD.x02180L
Like Mod.X02020L but provides 80 steps (1,25%2,5%5%7,5%10%12,5% up to 100%).
Tabletop or wheeled version available.
Optional instruments on request (Wattmeter, Voltmeter , Ammeter etc.)
Code 
Power 
Voltage

Steps 
MOD.3020L MOD.3021L 
0,3kVA 
230/400V 
7steps +0 20steps +0 
MOD.4020L MOD.4021L 
1kVA 
230/400V 
7steps +0 20steps +0 
MOD.5020L MOD.5021L 
2kVA 
230/400V 
7steps +0 20steps +0 
MOD.6020L MOD.6021L 
3kVA 
230/400V 
7steps +0 20steps +0 
MOD.7020L MOD.6021L 
5kVA 
230/400V 
7steps +0 20steps +0 
MOD.8020L MOD.8021L 
8kVA 
230/400V 
7steps +0 20steps +0 
MOD.9020L MOD.9021L 
10kVA 
230/400V 
7steps +0 20steps +0 
ADJUSTABLE CAPACITIVE LOADS  MONOPHASE  THREEPHASE
With three independent adjustment switch, one for each phase, for balanced and unbalanced steps.
MOD.x020C
Each phase can be loaded with 7 independent load values
plus the zero position. Using singlephase connection, it is possible to work with 21
different steps of regulation.
It is possible to build
balanced and unbalanced loads, singlephase or threephase Voltage: 220/380V (other voltages and powers upon request).
MOD.x02010C With 10 uniform steps; 3phase (star and delta). AC monophase provides 30 regulation steps.
MOD.x02015C With 12 loads, for 15 uniform steps; 3phase (star and delta). AC monophase provides 45 regulation steps.
MOD.x02020C With 20 uniform steps, (from 5% to 100%); 3phase (star & delta). AC monophase provides 60 regulation steps.
MOD.x02030C With 30 uniform steps, (from 3% to 100%); 3phase (star & delta). AC monophase provides 90 regulation steps.
MOD.x0201010C With 110 uniform steps, (from 1% to 100%); 3phase (star & delta). AC monophase provides 330 regulation steps.
With a single adjustment switch switch,
for synchronous variation of the three phases for each step.
MOD.x021C Like Mod.X020C but provides 20 steps (5%10%15%20%25%30% up to 100%) it can be used in Monophase or Threephase
MOD.x02140C
Like Mod.X020C but provides 40 steps (2,5%5%7,5%10%12,5% up to 100%).
Code 
Power 
Voltage

Steps 
MOD.3020C MOD.3021C 
0,3kVA 
230/400V 
7steps +0 20steps +0 
MOD.4020C MOD.4021C 
1kVA 
230/400V 
7steps +0 20steps +0 
MOD.5020C MOD.5021C 
2kVA 
230/400V 
7steps +0 20steps +0 
MOD.6020C MOD.6021C 
3kVA 
230/400V 
7steps +0 20steps +0 
MOD.7020C MOD.7021C 
5kVA 
230/400V 
7steps +0 20steps +0 
MOD.8020C MOD.8021C 
8kVA 
230/400V 
7steps +0 20steps +0 
MOD.9020C MOD.9021C 
10kVA 
230/400V 
7steps +0 20steps +0 
ADJUSTABLE RESISTIVE LOADS  MONOPHASE – THREEPHASE
With three independent adjustment switch, one for each phase, for balanced and unbalanced steps.
MOD.x020R Each phase can be loaded with 7 independent load values
plus the zero position. In singlephase connection is possible to work with 21
different steps of regulation.
It is possible to build
balanced and unbalanced loads, singlephase or threephase Voltage: 220/380V (other voltages and powers upon request).
Starting from 3kW it is equipped with an electric cooling fan.
MOD.x02010R With 10 uniform steps; 3phase (star and delta). AC monophase provides 30 regulation steps.
MOD.x02015R With 12 loads, for 15 uniform steps; 3phase (star and delta). AC monophase provides 45 regulation steps.
MOD.x02020R With 20 uniform steps, (from 5% to 100%); 3phase (star & delta). AC monophase provides 60 regulation steps.
MOD.x02030R With 30 uniform steps, (from 3% to 100%); 3phase (star & delta). AC monophase provides 90 regulation steps.
MOD.x0201010R With 110 uniform steps, (from 1% to 100%); 3phase (star & delta). AC monophase provides 330 regulation steps.
With a single adjustment switch switch,
for synchronous variation of the three phases for each step.
MOD.x021R Like Mod.X020R but provides 20 steps (5%10%15%20%25%30% up to 100%) From 0 to 480 Volt CA Monophase or Threephase and CC.
MOD.x02140R
Like Mod.X020L but provides 40 steps (2,5%5%7,5%10%12,5% up to 100%)
Code 
Power 
Voltage 
Steps 
MOD.3020R MOD.3021R 
0,3kW 
230/400V 
7steps +0 20steps +0 
MOD.4020R MOD.4021R 
1kW 
230/400V 
7steps +0 20steps +0 
MOD.5020R MOD.5021R 
2kW 
230/400V 
7steps +0 20steps +0 
MOD.6020R MOD.6021R 
3kW 
230/400V 
7steps +0 20steps +0 
MOD.7020R MOD.7021R 
5kW 
230/400V 
7steps +0 20steps +0 
MOD.8020R MOD.8021R 
8kW 
230/400V 
7steps +0 20steps +0 
MOD.9020R MOD.9021R 
10kW 
230/400V 
7steps +0 20steps +0 
Copyright © 1995  italtec Srl

